View : 165 Download: 0
Validation study of the Dinamap ProCare 200 upper arm blood pressure monitor in children and adolescents
- Validation study of the Dinamap ProCare 200 upper arm blood pressure monitor in children and adolescents
- Lee C.G.; Park H.M.; Shin H.J.; Moon J.S.; Hong Y.M.; Kim N.S.; Ha I.S.; Chang M.J.; Oh K.W.
- Ewha Authors
- SCOPUS Author ID
- Issue Date
- Journal Title
- Korean Journal of Pediatrics
- vol. 54, no. 11, pp. 463 - 469
- Purpose: To validate the Dinamap ProCare 200 blood pressure (BP) monitor against a mercury sphygmomanometer in children 7 to 18 years old in accordance with the 2010 International Protocol of European Society of Hypertension (ESH-IP2) and the British Hypertension Society (BHS) protocol. Methods: Forty-five children were recruited for the study. A validation procedure was performed following the protocol based on the ESHIP2 and BHS protocols for children and adolescents. Each subject underwent 7 sequential BP measurements alternatively with a mercury sphygmomanometer and the test device by trained nurses. The results were analyzed according to the validation criteria of ESH-IP2. Results: The mean (±SD) difference in the absolute BP values between test device and mercury sphygmomanometer readings was 1.85±1.65 mmHg for systolic BP (SBP) and 4.41±3.53 mmHg for diastolic BP (DBP). These results fulfilled the Association for the Advancement of Medical Instrumentation criterion of a mean±SD below 5±8 mmHg for both SBP and DBP. The percentages of test device-observer mercury sphygmomanometer BP differences within 5, 10, and 15 mmHg were 96%, 100%, and 100% for SBP, and 69%, 92%, and 100% for DBP, respectively, in the part 1 analysis; both SBP and DBP passed the part 1 criteria. In the part 2 analysis, SBP passed the criteria but DBP failed. Conclusion: Although the Dinamap ProCare 200 BP monitor failed an adapted ESH-IP2, SBP passed. When comparing BP readings measured by oscillometers and mercury sphygmomanometers, one has to consider the differences between them, particularly in DBP, because DBP can be underestimated. © 2011 by The Korean Pediatric Society.
- Appears in Collections:
- 의과대학 > 의학과 > Journal papers
- Files in This Item:
There are no files associated with this item.
- RIS (EndNote)
- XLS (Excel)
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.